AGILENT 33120A MANUAL PDF
for standard waveforms. The Agilent Technologies A Use the A to generate complex . Manual language options (please specify one). ABA US. For Safety information, Warranties, and Regulatory information, see the last page in this manual. Agilent A. Option Phase-Lock Assembly. oscilloscope or DMM and send it to your HP A for output. 3-year warranty. With your HP A, you get operating and service manuals, a quick reference .
|Published (Last):||9 January 2005|
|PDF File Size:||15.86 Mb|
|ePub File Size:||17.50 Mb|
|Price:||Free* [*Free Regsitration Required]|
Agilent Technologies 33120A User Manual
The can create waveforms represented by 4, 2 discrete amplitude levels. Page 31 Front-Panel Menu Operation Unless otherwise indicated, this manual applies to all Serial Numbers. Throughout this manual, the following conventions are used for command syntax. Notebook Asus A6RP repair manual service manual 1.
They are not included in the The carrier output can be controlled using either triggered or externally- gated methods. Chapter 3 Features and Functions Calibration Overview Calibration Message You can use the calibration message feature to record calibration information about your function generator. Page 7 Display Annunciators Adrs Function generator is addressed to listen or talk over a remote interface.
Back to home page. To replace the mAT fuse, order Agilent part number How do you get an MCU design to market quickly? PV charger battery circuit 4. During calibration, output amplitudes are calibrated for both the open-circuit voltage no load and the terminated output voltage loaded.
Chapter 5 Error Messages Execution Errors Settings conflict; amplitude has been adjusted The agklent output amplitude is not valid. Chapter 3 Features and Functions Output Configuration Output Frequency As shown below, the output frequency range depends on the function currently selected.
Agilent a output Disable – Discussion Forums – National Instruments
Dimension string array 80 elements Page This internal modulation signal is used to halt waveform memory addressing when the last data point is reached. The rate of frequency hops is controlled either by an internal source or from an external logic input. Tutorial The Agilent A is capable of producing a variety of signal waveshapes. How can the power consumption for computing be reduced for energy harvesting? It does allow you to set the output. Page Index points, number in arb waveform, RS shipping manua, 15 ports, serial COMarb download example, short form, SCPI commands, power-down recall,baud rate,agileent waveform, power-line voltage cables and adapters, single triggering, fuse, 16 configuration, FM modulation is enabled.
Agilent A service manual
Chapter 3 Features and Functions System-Related Operations System-Related Operations This section gives information on topics such as power-down recall, self-test, error conditions, and front-panel display control. Choose a web site to get translated content where available and see local events and offers.
Have one to sell? Arbitrary Waveform Overview The following is an overview of the steps required to download and output an arbitrary waveform over the remote interface.
SineSquare, Triangle, Ramp, or Arbitrary waveform. The burst rate frequency defines the interval between bursts. To Store The Instrument State Chapter 1 Quick Start To store the instrument state To store the instrument state You can store up to three different instrument states in non-volatile memory.
We have written the exercises in this chapter to prepare the function generator for use and help you get familiar with some of the front-panel operations.
The function generator outputs one burst or Single initiates one sweep each time you press the key. Page The A uses a bit phase accumulator which yields or approximately nHz frequency resolution internally. Input buffer overflow Output buffer overflow Command not allowed in local Minimum monthly payments are required.
Based on your location, we recommend that you select: Chapter 3 Features and Functions Calibration Overview To Change the Security Code To change the security code, you must first unsecure the function generator, and then enter a new code. Page Chapter 3 Features and Functions Calibration Overview To Unsecure for Calibration You can unsecure the function generator for calibration either from the front panel or over the remote interface.
The commands used for burst modulation are listed on page Set the carrier frequency between 10 mHz and 15 MHz kHz for triangle and ramp.