DATRON 4700 PDF
Model Specifications. Manufacturer(s). Datron. Instrument Model: Description: Autocal Multifunction Calibrator. Instrument Type(s). Calibrator. Application. Find great deals for DATRON AUTOCAL Multifunction Calibrator as Is. Shop with confidence on eBay!. Datron | Sale | Rental | Lease – Calibrator. Datron Calibrator. Contact us for pricing and availability. Description; Options.
|Country:||United Arab Emirates|
|Published (Last):||20 October 2004|
|PDF File Size:||10.70 Mb|
|ePub File Size:||6.58 Mb|
|Price:||Free* [*Free Regsitration Required]|
Voltage and frequency criteria are given in Section 2. The processor also controls the display, the IEEE Interface Bus and the operation of the ddatron and error circuitry. Next, the lit keys are man. A On Receipt from Patron: Set the DC Cmrent Sta.
Full text of “Datron: DATRON User”
Tliis can be used for recording, the results of verifying any of the three stored specifications. Connect measuring equipment 2-wire to HI and Lo Terminals, or 4-wire energizing current to! A Key LED lit signifies that conditions are valid for the selected operation, and not merely that the key has made contact.
The two modes may be selected together. These are compiled specifically for users without verification facilities, so that they can determine tile tolerance limits datronn the output, without reference to Section 6.
Wavetek/Datron 4708, 4707, and 4705 Multi-Product Calibrators
Procedural Information 7,35 7. Most interface communication tasks require sequences of coded messages to be sent over the interface. In all cases validity depends on using the same standard as reference, under the same conditions, dateon temperature.
After adjustment, the updated factors. Sum the Verification setup uncertainties and record m the Us column as voltage deviations- Refer to Appendix 5. Any combination of these two FAIL messages can appear in sequence, replacing the dxtron message. Four points are important: There is no Spec mode display on the front panel during remote operation.
The range of legends transmitted by the is ii. Before any calibration from the front panel is carried out, prepare the as follows: Introduction and a brief account of the 47700 design.
User Manual – Datron Instruments – Calibrator — Download free service manuals
Selection of High Voltage Daron The is capable of delivering LETHAL output voltages so program interlocks are used to ensure that users do not inadvertently select outputs in excess of 1. They can source direct voltage and current, alternating voltage and current, and resistance. Repeat procedure as required. Press Zero key to check its operation. Max Guard to Ground voltage of V rms 2. The correct procedures for making the perform for you. Any five frequency values within the range of the instrument can be stored in volatile memoiy.
Poor Maintenance — Over time or in dusty working environments dust and debris can build up on the filter and inside the unit. Specifications — what accuracy you can expect to get from your instrument. The labelled facility is not datrkn. Display and Key Checks If the Test Key is re-pressed before pressing any other key, a visual sequence tests the front panel: As an aid to the controller and printer, the DMM can send another message with the last byte to be transferred EOI-end or identic, using another bus management line.
The millivolt range is then verified at this standardized value by DVM measurement. Specification Toierance Per unit – P codes.
Ensure that the DVM is still connected to the terminals as shown adtron. In this condition, a. This status report is also available manually by pressing Error key then Lim—.
Use conductors, joints and terminals with a good margin of current-carrying capacity. RANGE key value is tiom. Ensure that the is set to 470 1 00m V AC range. This can increase internal operating temperatures affecting reliability.
Set the IVD ratio to 1: Activation of Commands — what the does with the commands it receives.