HIOKI 3284 MANUAL PDF
A Complete HIOKI Digital & Analog Clamp Tester /A AC .. CARRYING CASE ×1. Hand strap ×1. Instruction manual ×1. Hioki Pdf User Manuals. View online or download Hioki Instruction Manual. View and Download Hioki instruction manual online. Clamp On AC/DC HiTester. Measuring Instruments pdf manual download.
|Published (Last):||28 October 2008|
|PDF File Size:||3.48 Mb|
|ePub File Size:||1.12 Mb|
|Price:||Free* [*Free Regsitration Required]|
AC/DC Clamp Meter | Clamp On AC/DC HiTester – Hioki
We have tried to bring this manual as close to perfection as we could achieve. If perchance you find any unclear portions, mistakes, omissions, or the like, we would be most obliged if you could please notify us of them via any HIOKI agent, or directly. In particular, check the accessories, panel switches, keys, and terminals. If the unit is damaged, or fails to operate according to the specifications, contact your dealer or HIOKI representative.
Check the Unit and the Supplied Accessories However, mishandling during use could result in injury or death, as well as damage to the instrument. Indicates that incorrect operation presents an extreme hazard that could DANGER result in serious injury or death to the user. In the manual, the symbol indicates particularly important information that the user should read before using the instrument.
These are defined as follows. Secondary electrical circuits connected to an AC electrical outlet through a transformer or similar device. Using a measurement instrument in an environment designated with a higher-numbered category than that for which the instrument is rated could result in a severe accident, and must be carefully avoided.
CLAMP ON AC/DC HiTESTER
Chapter 2 Measurement Procedure Explains how to use the for measurement. Using a one-chip microprocessor, the There isn’t an annunciator ” “. FAST Instead, the unit symbol blinks.
Reset all the data in a REC function. Select a proper mode according to the waveform shown below: Remove the rear cover and insert a battery. Refer to “Chapter4 Battery Replacement”.
Press to turn the unit on. Then the model name is shown, and the bar graph indicates the battery condition. If this happens, perform a zero-cancel correction 2. Open the top ends of the clamp core and clamp the measured conductor so that it passes through the center of the clamp core. Press and select a measurement mode for the measured circuit.
The measurement mode is switched as follows. Quit the peak measurement mode, and perform the auto-zero adjustment by.
HIOKI INSTRUCTION MANUAL Pdf Download.
Set a range based on the unit’s measurement range and other instruments, such as recorders. Manua conversion table for ranges is provided below. Figs 2 and 3 As the measured value is small to the range response time becomes long.
Press to display 2. Slide the slide cover up using the slide knob. Next, insert the red test lead to V and the mannual test lead to COM of the voltage measurement terminal. If the current range of the measured circuit is known, set the current range to the manual range.
The auto power-off function is automatically disabled.
Always switch to the range you will use a manual range before performing the adjustment. The counter update may be altered according to measurement conditions. Every time one minute elapses, one segment of the flashing bar graph goes on.
Hioki 3284 Instruction Manual
When all segments on the bar graph go on, the elapsed time is 30 minutes. While the recording function is being deactivated, data is not updated, even if the clamp sensor is disconnected from the conductor. If the function is activated when there is no input, the minimum value will remain zero. Also, when deactivating the recording function, press the key to terminate HOLD Replace the battery to ensure accuracy.
The torque about 0. Remove the two fastening screws of the rear cover, using a Phillips screwdriver. Remove the rear cover. It is recommended that a new battery should be used, when the blackout seems to be short. Symptom Battery Battery clip Test leads Unit does not come indication appears and unit immediately turns off. Unit turns off during use. A smaller value will be displayed for a high inverter carrier frequency. Auto-zero adjustment function and 2.
Pack the unit carefully so that it will not be damaged during transport, and write a detailed description of the problem. HIOKI cannot bear any responsibility for damage that occurs during shipment.
Also See for Instruction manual – pages. Page of 86 Go. Table of Mabual Add to my manuals Add. Page 5 We have tried to bring this manual as close to perfection as we could achieve. Shipping Check In particular, check the accessories, panel switches, keys, and terminals. Comments to this Manuals Your Name. Enter text from picture: By hikki ‘print’ button you will print only current page.
To print the manual completely, please, download it. Don’t show me this message again. Got it, continue to print.
Don’t have an account?