ISO 7886-3 PDF
Objetivo: ISO specifies the properties and performance of sterile single-use hypodermic syringes with or without needle, made of plastic materials . Find the most up-to-date version of ISO at Engineering Sterile hypodermic syringes for single use – Part 3: Auto-disable syringes for fixed -dose immunization (ISO ); German version EN ISO
|Published (Last):||1 November 2006|
|PDF File Size:||10.75 Mb|
|ePub File Size:||20.24 Mb|
|Price:||Free* [*Free Regsitration Required]|
Increase the force to a maximum of N or until the plunger is capable of delivering a second dose from the same syringe. It should not be possible to defeat the auto-disable feature by isso and re-inserting the plunger. Do you want to buy this document? The piston shall not become detached from the plunger when tested in accordance with Annex B oso ISO If you need help with the purchase or have any queries please email enquiries standards.
NBN EN ISO | NBN
This part of ISO does not specify the design of the auto-disable feature, which is left to the discretion ofthe manufacturer. When the fiducial line of the piston coincides with the zero graduation line, the preferred minimum length of the plunger from the surface of the finger grips nearer to the push-button shall be 8 mm. Sterile hypodermic needles for single use ISO For syringes with integrated needle, 14,2 of ISO For the purpose of deciding whether a particular requirements of this standard is complied with, the final value, observed or calculated expressing the result of a test or analysis, shall be rounded off in accordance with IS 2: Buy Do you want to buy this document?
Details of which is given in National Annex A. Figure 1 — Schematic representation of auto-disable syringe for fixed dose IS Part 3: No secondary or additional action on the part of the user shall be required. IS Part 3: For dated references, only the edition cited applies. The plunger should be of a length adequate to allow the piston properly to deliver the designated fixed dose.
Connect the needle of the syringe to the rubber stopper of the tubing of the reservoir B.
МКС EN ISO 7886-3:2011
Autodisable syringes for fixed-dose immunization ISO Although we endeavour to maintain an up-to-date catalogue, on occasion we may not have the latest version of a document, including where it has not yet been 77886-3 through an international feed.
Re-use of injection equipment in the absence of sterilization has increasingly let to transmission of blood-borne pathogens this standard.
Both the needle and the syringe shall be rendered incapable of re-use after delivery of the intended fixed dose, under normal conditions of use. It does not address compatibility with injection fluids.
Then click on “Details” and select the isl version. This part of ISO is not applicable to syringes for use with insulin specified in ISOsyringes madeof glass specified in ISOsyringes for use with power-driven syringe pumps specified in ISO ,auto-disable syringes for variable dose delivery and syringes designed to be prefilled.
For undated references, the latest edition of the referenced document including any amendments applies.
This part of JSO is not applicable to syringes for use with insulin specified in ISOsyringes made of glass specified in ISOsyringes for use with power-driven syringe pumps specified in ISOauto- disable syringes for variable dose delivery and syringes designed to be prefilled.
NOTE Annex B gives a suggested test method and performance criteria for the forces required to move the plunger. Syringes for manual use IS Review of Indian Standards Amendments are issued to standards as the need arises on the basis of comments. The projection of the plunger and the configuration of the push-button should be such as to allow the plunger to be operated without difficulty.
Summary This part of ISO specifies the properties and performance of sterile single-use hypodermic syringes withor without needle, made of plastic materials and stainless steel and intended for the aspiration of vaccines orfor the injection of vaccines immediately after filling.
This standard is intended to cover ‘fixed dose’ immunization syringes that are rendered inoperable after delivery of the intended dose. Using the test device C. This part of ISO does not specify the design of the auto-disable feature, which is left to the discretion of the manufacturer.
The corresponding Indian Standards which are to be substituted in their places are listed below along with their degree of equivalence for the editions 7886- International Standard ISO Upon delivering a fixed dose of vaccine, the syringe isautomatically rendered unusable. You will be redirected to our e-shop where you can buy the document. Adjust the relative positions of the syringe and reservoir so that the water level in the reservoir is lso level with the mid-point of the syringe barrel.
Connect the nozzle of the syringe to the tubing of the reservoir B.